Tektronix DP501 Prescaler 1.3 GHz
|Back to Tek TM500 Series Summary Page|
Quick Specifications / Photos|
Misc. Part Numbers
|Orig Cost||Low Bid||High Bid||average|
no sale ($82 < reserve) 8/10/01 #1624260692 dogchow (with DC503A)|
$100.00 8/8/01 #1622904090 lyntec
$84.88 12/1/02 #1920821905 j.a.
$82.11 2/2/03 #3308632542 dwconvery (works) (13 bids $25 start)
$87.00 (+$10 ship) 5/3/06 #7614061565 onlinesurplus (7 bids $4.99 start) (clean-untested)
Two operating modes, prescale and direct, are available, In the prescale mode, the DP 501 divides the input frequency by 16. In the direct mode, the input of the prescaler is connected directly to the input of the counter. Selection of operating modes is accomplished by a front panel pushbutton or by the PRESCALE commend sent via the IEEE 488 bus to the microprocessor in a programmable counter. Communication from the IEEE 488 compatible counter to the DP 501 is accomplished by a one-wire connection through the rear interface connector.
For certain counters with accessible clocks and time bases, the counters time base signal may be routed to the DP 501, divided by 16, and returned to the counter. In these cases, the counter displays the actual input frequency of the DP 501 (some display flicker is present when using the DC 504A). Where these connections are not possible, the counter reading must be multiplied by 16 to obtain the correct DP 501 input frequency. In the prescale mode and automatic attenuation circuit provides optimum immunity to input noise.
In the direct mode the input signal to the DP 501 is looped through the input and output terminals to the counter in a 50 Ω environment.
An interface is provided for operation of the counter, spectrum analyzer, and tracking generator (or other suitable device) for accurate readout of a selected frequency. This interface is implemented through the rear interface connector and is useable only with counters having an arming input.
above excerpts from DP 501 DIGITAL PRESCALER Instruction Manual
Connect the OUTPUT bnc connector to the input of the counter using a 50 Ω coaxial cable. A 50 Ω termination may be used at the counter input. If a termination is used, the output of the DP 501 will be terminated in both prescale and direct modes. Connect the signal source to the input connector using a 50 Ω cable.
A FET probe such as the Tektronix P6201 can be used to input signals to the DP 501. The DP 501 may be operated in either the prescale or direct mode using the FET probe. A 50 Ω environment for the DP 501 output connections must be maintained in the direct mode to insure proper FET probe termination.
When the input signal exceeds the maximum power level (approximately 20 dBm) the OVERLOAD light illuminates and the input circuit opens. The overload disconnect capability does not operate in the direct mode. The OVERLOAD light remains illuminated until the overload is removed (input signal lowered to 100 mV, -7 dBm) and the RESET button is pressed or PRE ON command is sent over the GPIB through a programmable counter. If the RESET button is pressed with the overload at the input, the unit cycles to prescale and immediately to overload. If the DP 501 is turned on with the input overloaded the DP 501 switches to the direct mode. If the button is pushed with the overload present at the input the instrument goes into the overload mode. If the input frequency is high enough (about 1 GHz) the DP 501 remains in the overload condition (whether the RESET button is pressed or not) until the input overload is removed. Approximately 500 ms is required for the overload circuitry to recover after the overload is removed.
above excerpt from DP 501 DIGITAL PRESCALER Instruction Manual