Bandwidth : DC to 1 GHz|
Risetime : <350 ps
Attenuation : 10X
Input Resistance : 10 MΩ
Input Capacitance : approx. 1.9pF
Scope Input Resis : 50 Ω
Readout Pin : Yes
Ground Ref : ?
ID Button : Yes
Max Rated Voltage : 40 VDC
Out of Production 11/1/2000
Designed for the 11000-Series Oscilloscopes and plug-ins
The P6204 and P6203 high-performance Active Probes round out the capabilities of the 11000-Series
oscilloscopes and plug-ins.
The P6204 is a 1-GHz, high impedance (high input R, low input C), miniature size, 10X Active (FET) probe
intended for use in high speed digital and analog circuitry. The variable dc offset voltage (±10V) is
adjustable via mainframe controls, allowing offset of any dc component within the range of the control. This
allows signals with dc components to be brought into the probe's dynamic range.
Although primarily designed for the 11A52 and 11A71 plug-ins, the P6204 may be used with the other 11000-Series
The 50-ohm termination required by the P6204 is automatically accommodated by the 11000-Series mainframe. When
an 11000-Series active probe is connected to an input connector of an 11A32, 11A33, or 11A34 Amplifier plug-in
unit, the plug-in input impedance is automatically switched to 50 ohms.
The P6204 Active probe has an IDENTIFY button which, wnen pressed, will cause any one or more of
several actions to be taken by the mainframe; these include autoset, recall the next in a series of stored
setups, invoke the automatic-measurement function, or issue an SRQ. These actions are selectable via the
mainframe Utility Menu.
The P6204 Active Probe provides communications (intelligent probes) through the TekProbe
interface connector. The probe can (if available on the specific probe) communicate one or more of the
following: it's type, serial number, attenuation ratio, input impedance, offset scale factor, and other
key parameters to and from the host instrument. Power for the Active probes is also supplied through the
TekProbe interface, eliminating the need for extra cabling and external power supplies.
The P6204 may also be used with other oscilloscopes bia the 1103 TekProbe power supply.
(excerpt from 1989 Tektronix catalog.)
|Attenuation :||10X ±1.5% into a 50Ω ± 0.5% load.|
|Input Resistance :||10 MΩ|
|Input Capacitance :||1.9 pF. (Typically 1.7 pF)|
|Bandwidth (-3 dB) :||DC - 1.0 GHz (1.1 GHz Typical)|
|Risetime (Probe Only) :||<350 ps (<320 ps typical)|
|Risetime (System with 11402 & 11A71) :||<490 ps|
|Maximum Input Voltage :||40 VDC
|Weight :||Approximately 680 g (1.3 lbs) including accessories|
CABLE : 1.5 m (60 in)|
Probe will operate within specifications over the following ranges:
Temperature : 0° to +50°C (+32° to +122°F)
Humidity : Five cycles (120 hr.) at 90% to 95% relative humidity.
Altitude : To 15,000 feet.
(excerpt from P6204 Instruction Manual 070-6949-03)